DomDV Posted May 24, 2019 Share Posted May 24, 2019 Hi My Openscope MZ board arrived yesterday, and I've been setting it up following the video series. My board seems to be displaying some issues: 1: In Waveforms live, The Waveform generator does not display a waveform. I select the power button icon on the wavegen, then select a channel in the same way and hit Single. Nothing happens. No trace, or waveform. 2: I have the board set up with a direct usb connection and a wifi connection across the network however, the blue led wifi indicator on the openscope does not stay lit, and regardless of which device connectivity I choose the result is the same ie no waveform generated in waveforms live. 3. Occasionally when selecting "Single" I also get a message saying that "the device is already in use" when its not measuring anything. 4. After restarting the device, I attempted to see a waveform again and this time I hit "Single" and got a message saying "The AWG may have been running and has been stopped, please try again". I tried again, hit "single", nothing happened, so I hit "Abort" and got this message "Error stopping trigger, if this problem continues, restart the device and waveforms live". I tried again, and got nothing again. The experience so far has been frustrating. Is my board faulty or not configured correctly. I'm at a loss right now. Any help would be very much appreciated. Link to comment Share on other sites More sharing options...
Create an account or sign in to comment
You need to be a member in order to leave a comment
Create an account
Sign up for a new account in our community. It's easy!Register a new account
Already have an account? Sign in here.Sign In Now